Anyone experiencing problems during your treatment where your arterial pressure will jump all of a sudden and then you get the red 24? Then it could be good for a while and happen again.
This seems to happen right after raising the blood flow. It eventually comes down to normal, but sometimes requires a pod reset.
I called tec support and they stated this is a know issue happening sporadically with certain cartridge lot #'s. There are a few options you have…
- reset the pressure pod
- back down on the blood flow approx 50 points and see if the pressure goes back down, then gradually go back up.
- could be the pressure pod line where it attaches to the machine. Could be too tight, or not tight enouph.
I was told that If this happens, pease report it to Tech Support to the issue can be set on a higher priority.
One known lot number is 9077717. Please post any other lot numbers anyone may be aware of.